Environmental Test Chamber Installation
Remote Refrigeration or Remote Air-Cooled Condenser
If the chamber has a remote refrigeration section or remote air-cooled condenser, determine where these components will be located. Is there unobstructed space for interconnect routing? Routing of refrigeration and electrical interconnect above drop ceilings and through an obstructed area increases installation costs and decreases efficiency. Try to locate the equipment in an area with open access for interconnecting the components.
Panel Walk-In Chambers
Floor quality is very important when installing a panel Walk-In Chamber. The floor must be level to a standard of 1/8” per 10 feet or better in any direction. If the floor the chamber is to be built on has high and low points it will not properly support the chamber structure. It also makes it very difficult to assemble to structure squarely. This creates problems with the structure remaining sealed properly and will reduce the life of the panels.